DATAQ DI-158U DRIVER DOWNLOAD
ChannelCount Reads the number of currently enabled channels Variable: Please note the front-end circuit of DI has a low-pass filter and it may not respond ideally for high frequency signals. MatchSerialNumber if this is set. Original channel count is unmodified PostTriggerLength should be zero if all data points above the level should be captured, or one if only the first one should be captured PreTriggerLength integer specifies the number of scans before the trigger point, 0 to scans can be specified default: One frame is equivalent to one sample from every active channel. Failed to create monitoring process, please poll the data from VB Timer function
|Date Added:||20 July 2016|
|File Size:||28.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
DataQ Instruments Starter Kits
For most C programmers, a variant of integer array returned by GetData is not what they would like to ci-158u. PChannel index changes P rogrammable Channel allocation table, or scanlist if you are using Dataq’s term.
One frame is equivalent to one sample of each active channel. SetLimitLine int type, int threshold specifies the type of monitoring and threshold to be used in LimitLine operation.
Fired when an error is detected by the control. To convert the AD Readings, check here.
You cannot use PChannelPGain property to change the channel and gain allocation table. Unsupported trigger channel If you try to access the digital channel which is not available on the device, the result is meaningless see below for further explanation.
WinDaq Data Acquisition Software
When you specify a sample rate, an attempt is made to report data at that sample rate. For example, if you specify “” as the integer for EventLevel, the NewData event fire whenever or more new data points have been gathered. GetDataEx returns waveform data to an integer array as binary data. MatchSerialNumber if this is set.
PreTriggerLength should be zero PostTriggerLength should be zero if all data points above the level should datxq captured, or one if only the first one should be captured AvailableData: CommPort Reads serial port Variable: When this is not possible, the data will be reported at the closest achievable rate to the sample rate you requested. Specify the Dataq serial data acquisition device model number you are going to use or find out which one you are currently using.
GapInDataStream indicates a gap is detected in the data stream. The default di-18u for gain di-158 is dwtaq. CommPort outside of range, which are from 0 to 64 for physical comm port, up to for USB-based virtual comm port.
Dataq Instruments Di-158u USB Data Acquisition Starter Kit
All It serves as the entry point for the superkeys of site license Example: We highly suggest DataqSDK. For DI and DI, the rate channel has gain control. Failed to communicate to the device Unsupported trigger mode ChannelCount Reads the number of currently enabled channels Variable: Please note the front-end circuit of DI has a low-pass filter and di-18u may not respond ideally for high frequency signals.
Specify the COMM port serial port you want to use or find out which one you are currently using. Channel 8 is always digital input, Channel 9 is always Rate input, and Channel 10 datq always a bit Counter input. Count is an integer and defines the number of scans you are requesting.
The data will be aligned with the first channel. When a specified PChannel passes the limit line, event LimitLine will be fired.
There are multiple functions for this property. All Valid Acquisition Mode: Both bit and bit Windows runs bit applications properly. When Ultimaserial is first loaded, the channel allocation table is 1, 2, 3, The minimum throughput rate for other 1xx is: